|Name:||High Temperature Pressure Transducer||Pressure Ranges:||0-2000bar|
|Overload Pressure:||200%FS||Output Signal:||4~20mA, 0~5V, 0~10V, 1~5V,0.5-4.5V|
|Accuracy:||0.1%FS, 0.25%FS, 0.5%FS||Usage:||Test Liquid And Air|
|Supply Voltage:||12~30V||Operating Temperature Range:||-40~+85°c|
0.25%FS High Temperature Pressure Transducer,
0.1%FS High Temperature Pressure Transducer,
0.1%FS high temperature pressure transmitter
0.1%FS 0.25%FS High Temperature Pressure Transducer Test Liquid And Air
Description of High Temperature Pressure Transducer
High Temperature Pressure Transducer use Sputtered film technogy to test liquid and air pressure .
The High Temperature Pressure Transducer is fully welded with intrinsic safety and explosive-proof futures, so can by applied in harsh industrial environments and high temperature environments like 0 °c - 150 °c~230°c
High accuracy and stable performance got very good feed back from customer.
Feature of High Temperature Pressure Transducer
● 150 °c~230°c Operating temperature
● Sputtering film pressure cell
● Fully weld
● CE certificate
● Intrinsic safety, explosive-proof
● High accuracy
Specifications of High Temperature Pressure Transducer
|pressure medium||gas liquid compatible stainless steel|
|output signal||4~20mA, 0~20mA, 0~10mA ,0~5V, 0~10V, 1~5V|
|accuracy||0.1%FS, 0.25%FS, 0.5%FS|
|load resistance||RL = (U-10)/0.02(Ω) (4~20mA current output), ＞20KΩ (voltage output)|
|long term stability||0.25%FS/year|
|compensated temperature range||-20~120°c|
|operating temperature range||-50~+150°c (typical), -50~+230°c|
|storage temperature range||-50~+170°c|
|temperature coefficient of zero||0.3%FS/10°c|
|temperature coefficient of span||0.3%FS/10°c|
|insulation resistance||>100MΩ@ 50VDC|
|process connection||G1/4 or others|
|electrical connection||DIN 43650 or others|
|material of wetted part||1Cr18Ni9Ti|
|Elect strength||500V@60 second|
|material of housing||1Cr18Ni9Ti|
1.Q: How long is the delivery?
-- Mostly within 10days
2.Q:Can I get a sample?
-- Sure, contact our sales please
3.Q:What is your MOQ?
4.Q: How about warranty time?
Contact Person: Jason